Hello C S!
C S Miller wrote:
> One simple question: is there some way to replace the BusLogic
> driver with the correct one in VMWare? (And does it really
> matter? I don't think AdvanSys cards are necessarily
I think it does not matter. It should work fine. See below.
> Or is there something going wrong in the pClamp software
> itself? Every time I fire up my recording protocol, it
> immediately shuts down with the log message: "No data was
> recorded and the empty data file blah-blah.abf has been
> deleted." Might there be something simple I am overlooking
> that could cause this?
Could it be that you are not triggering? To check change
aquisition mode to "gap-free" ("Aquire" -> "Edit Prot.") and make
sure that "start trial with" in "trigger" is set to "immediate".
If it doesn´t work, you could call Ward from Axon Instruments. I
don´t think it´s a problem with your VM since calibration works
fine... If I can be of any more help send me an e-mail (don´t
forget to manually remove "-remove-").