I often get sharp spikes in my ABI 310 electrophoresis chart which upsets
the baseline and can take some time to get back to normal. Any idea what
could be causing this? The only thing we have thought of are power surges
from the mains line. It happens often, so it may be some other machinery we
have running at the same time.
Get Your Private, Free Email at http://www.hotmail.com